· 10. The straight handle is installed on the collet, and the stator is speculated by finger light, so that the eccentricity is 0.5mm and the speed is 400-600 / min, so as to measure the stator
· If you want to touch the end face of the workpiece, the touch movement of the dots is that the detector is not vibrating, just like the static state.
· Then touch with inching feed, the measuring stator will slide in one direction, and the sliding starting point is the required reference position.
· It is not suitable for horizontal meter.